YP Magnetic HLY10-100PMMA 1-Axis Helmholtz coil/Magnetic field coil/Magnetic Field Generator
Get Latest Price
Place of Origin:
YP Magnetic Technology Development Co., Ltd.
Manufacturer, Trading Company
On-time delivery rate
Magnetic Field Curve
Power Supply Cable
OPTIONAL PURCHASE LIST
Manual Lifting Sample Workbench
Manual Rotating Sample Workbench
The founder Dr. Yupu Zhang and his team
Since 1956, Dr. Yupu Zhang and his research team have been engaged in magnetic measurement research in the laboratory of Jilin University, and in 1985, they developed the First Vibrating Sample Magnetometer in China and won the Third Prize of National Science Progress Award.
Officially established in 2002, YP Magnetic Tech-nology Development Co., Ltd. has become the most influential manufacturer which specializes in magn-etic field generation instruments and equipment in China. This company supplies hundreds of magn-etic-field-related experiments instruments and devices for the laboratories in China and abroad every year.
After decades of development, this company has been pursuing technological innovation and brea-kthrough, with the most advanced capabilities of calculation, simulation and processing in the area of design of magnetic field generating apparatus and equipment and magnetic measurement prod-ucts.
* Helmholtz Coil Model No.: HLY10-100（PMMA）, under the excitation of the exciting power source, the homogeneous magnetic field is produced in the central region of the coils, and there is the linear relation between the current and the magnetic field, and the magnetic field direction can be designed to be horizontal or vertical.
* DC resistance at 20 ℃: About 2.6 Ω; * Insulation resistance: Better than 10 MΩ.
* Natural Cooling
Dimension and Weight
* Coil Nominal Diameter: 200 mm, Inter-Coil Spacing: 53 mm * Product Weight: About 9.5 Kg
Magnetic Field Parameters
* Center Magnetic Field Strength: 10 mT @ 5 A, B/I coefficient: 2 mT / A * Within the center volume of S φ 50mm, homogeneity is better than 1 %. * Within the center volume of S φ 28 mm, homogeneity is better than 0.1 %.